Trigger Philips Achieva
I hope you are ok, i´m still having problems with the trigger. I tried the option with the internal swiches (put both down), and i changed the wide of pulse (directly of the options of the scanner) to 200 us, 500 us, 1 ms and 5 ms, and it couldn´t function ok. When i changed it, i obtained some range of pulses separate by 1 dynamic, but they were not sequencially like i programed it (each dynamic, beginning in the firts dynamic),and in many times i didn´t receive any pulse.
I configured the controller with 9600 bd, optical isolation ON, internal swiches down, Lumina device.
We are really worried about it , I appreciate if you can contact some in the world who has been working with Lumina and Philips Achieva (or just Philips scann) and has had success with the trigger. We really need this information, i don´t need to contact these people directly, but thought you i think that it is possible.
Please give our mail to these persons if is necessary to make a contact:
I appreciate your help and your time,
Thanks in advance,