We recently got a lsc-400b Lumina equipment, which is connected to a HDX GE MRI machine, and to a laptop with presentation software (v13). We try for different mri sequences but we could only read a couple of pulses. Lsc-400 was connected to the r15 port of the resonator using a trigger lumina cable for GE. could anyone give us some advice on this? (we check for the dip switch and both were top).
Any feedback is appreciated